Induction touch panel
The LCD display control box can be rotated in horizontal direction.
Adjustable height and position of the placement of the suspended wrench is suitable for different specifications of wrenches.
The axis is designed with effort saving mechanism that makes high torque examination easier than usual.
Feature with aluminum alloy base, which is light and easy to carry.
Length x Width x Height (cm): 82 x 52 x 48 (The 30cm monitor hasn’t installed yet)
Measuring range: 40~400Nm
Accuracy: Reading +/-1%
Unit selections: Kg-cm, in-Lb, N-m and Ft-Lb.
Data transmitting function: The measurements will directly be sent and recorded into an Excel file on your computer.
Simple self-calibration: There are, except the zero point, 4 calibration points, and 5 setting modes of calibration points. Furthermore, the zero point is the commonality of different modes.
Press once can return the highest torque value of the last measurement to zero, then a new term of calibration can be implement.
This specific button can execute 2 movements. To operate program setting, press once. When encountering lists with numbers, press once to add 1 to the value.
This specific button can execute 2 movements. Press once to go to next page when entering the menu. When encountering lists with numbers, press once to minus 1 to the value.
Press to enter the setting or confirm the parameters.
This is the power button of the Professional Torque Tester. Press once to turn on the machine, and the content will be displayed. Press again when finish the calibration process, and the power and display screen will shut down.
Different from the old model that has only 1 mode of 4 calibration points, the Professional Torque Tester is designed with 5 modes that allows users to set 5 sets of different values at one time and select the mode they require, which can speed up the selecting process of picking different specifications of torque wrenches.
There are 5 calibration points can be set in 1 mode according to the value you require. Due to the unchangeable zero point, only 4 points are illustrated in the graph.
When operating the calibration, the calibration rod should maintain accurate.
Set the torque tester＞Set the torque wrench＞Place the torque wrench onto the torque tester to operate the tripped test.